Our products are made using the following:
Stainless steel frame:
The frame of the test bench is fabricated by making use of qualitative stainless steel that ensure sturdiness and longer service life.
Provided hydro lever is a pressure lever used for conducting pressure tightness test f or instance 1.6Mpa and 2Mpa
These are provided to check pressure of the pump
On/Off valves have been installed for conducting the test on the water meters
Meter locking cylinder:
These are installed to lock the meter or meter tightness
High grade temperature indicator is provided to check the temperature of water. As per the IS779, temperature should not less than or equal to 450C
These gauges check pressure of line and hydro efficiently
Fabricated by making use of supreme quality 2.5mm thickness sheet of stainless steel these water stabilizers remove all the germs of water. The capacity of the water stabilizer is 1000 liters.
Removes existing and prevents future build-up
Removes existing corrosion and prevents rust
Controls pitting due to entrained gases
Eliminates unpleasant odors
Eliminates the need for treatment chemicals
Removal of calcium deposits
An increase in water volume and protection of plumbing from corrosion
Prevention of scale and corrosion
Water level indicator
It checks the level of water, which should be at its maximum stage with the help of 23.
To check and control the pressure in the stabilizer.
A panel board with three switches (red, green and yellow) have been provided for operation. While starting the machine red light blinks which indicates that the machine is getting supply, and when red light turns into green it means that machine is ready to be operated. If any fault occur such as overload, low voltage, reverse rotation of the pump or any kind of electrical deficiency yellow light starts blinking.
These on/off valves are used to increase the pressure which is needed to test bigger size meter otherwise it should be in the on mode
Measuring vessel (0-20l):
These vessels checks the quantity of water which had passed through the water meter i.e. 0-10-20l.